Internal Halogen-quenched, uncompensated GM tube with thin mica window, 1.4-2.0 mg/cm2 areal density. Effective diameter of window is 45 mm (1.77 in.). Radiation symbol on front label and end panel marks the center of the detector.
mR/hr - .001 (1µR) to 100 CPM - 0 to 350,000 µSv/hr - .01 to 1000 CPS - 0 to 5000 Total Counts- 1 to 9,999,000 counts Accuracy (Referenced to Cs137) Typically ±15% from factory, ±10% with NIST Source Calibration
Detects Alpha down to 2 MeV. Detects Beta down to .16 MeV; typical detection efficiency at 1 MeV is approx. 25%. Detects Gamma down to 10 KeV through the detector window. 3340 CPM/mR/hr (137Cs). Smallest detectable level for 125I is .02 µCi at contact.
mR/hr .001 - 50 and CPM 1 - 160,000. 70db @ 1m. Pulsating beeper sounds the alert. Adjustable alert levels are used for mR/hr / CPM, and µSv/hr / CPS. Alarm will sound when in Timer Mode when set alarm threshold is reached.
Meter will hold at OVER RANGE in fields as high as 100 times the maximum reading.
Display updates every 3 seconds. At low background levels, the update is the average for the past 30-second time period. The timed period for the average decreases as the radiation level increases. Display
Backlit 4 digit liquid crystal display with indicators.
Red LED flashes with each radiation event.
Internally mounted beeper (can be switched off for silent operation)
3.5mm jack for CMOS or TTL devices and counts to computer or datalogger. USB for use with Free Observer USB Software for PCs
One 9-volt alkaline battery (non-rechargable) or USB-AC Adapter. Battery life is approx. 1200 hours at normal background.
-10ºC to 50ºC (14ºF to 122ºF)
273 g (9.6 oz.) without battery.
150 x 80 x 30 mm (5.9 x 3.1 x 1.2 in.).
Carrying Case, Certification of Conformance, USB Cable, Observer USB Software